Signal-to-Noise Optimization. Noise Sources Most Commonly Encountered 1. Detector Noise 2. Shot Noise 3. Flicker Noise. Detector Noise. Associated only with the detector, and therefore constant for a given set of detector conditions. N detector = Constant (S/N) det S. Shot Noise.
Download Policy: Content on the Website is provided to you AS IS for your information and personal use and may not be sold / licensed / shared on other websites without getting consent from its author.While downloading, if for some reason you are not able to download a presentation, the publisher may have deleted the file from their server.
Noise Sources Most Commonly Encountered
1. Detector Noise
2. Shot Noise
3. Flicker Noise
Associated only with the detector, and therefore constant for a given set of detector conditions.
Ndetector = Constant
Noise associated with the random transfer of electrons across a p-n junction.
Ex: Whether or not a single photon falling on a detector will actually produce a signal.
S = measured signal
e = charge on electron
Δf = frequency bandwidth
Shot noise is usually the limiting source of noise near the detection limit
Random noise with a 1/f frequency dependence.
f = sampling frequency
Flicker noise includes slow drifts in signal intensity caused by such parameters as temperature, flow rates, etc.
where ξ = flicker factor (unit-less)
(S/N)fl = S/ξS = 1/ξ
ξ 1/f so (S/N)fl f
and f = frequency of data collection
(S/N)fl = 1/ξ
determine the slope (m)
1. m = 1 → Detector Noise
2. m = ½ → Shot Noise
3. m = 0 → Flicker Noise
When we measure N experimentally, it is often a combination of all of the noises present in the system. The preceding equations are useful to determine which type of noise dominates in a certain situation.