Federal Aviation Administration. Radiant Panel Update. Pat Cahill AJP-632; Fire Safety Team Wm. J. Hughes Technical Center Federal Aviation Administration.
Download Policy: Content on the Website is provided to you AS IS for your information and personal use and may not be sold / licensed / shared on other websites without getting consent from its author.While downloading, if for some reason you are not able to download a presentation, the publisher may have deleted the file from their server.
Pat CahillAJP-632; Fire Safety TeamWm. J. Hughes Technical CenterFederal Aviation Administration
The last two panels we received were out of specification (found this during the 3-position check and the “very high”set point temperature)
Both panels were sent back to our distributor who in turn, sent them back to Watlow in St. Louis
Watlow has informed us that they have replaced the emitter strips on both panels
It is imperative that a 3-position check be performed when installing a new panel
Labs should work with their distributors if problems are foundRadiant Panel