Loading in 2 Seconds...
Loading in 2 Seconds...
Fresenius Medical Care Peritoneal Dialysis Connections. Safe-Lock to Baxter Connection. Connections using Fresenius Medical Care products.
Download Policy: Content on the Website is provided to you AS IS for your information and personal use and may not be sold / licensed / shared on other websites without getting consent from its author.While downloading, if for some reason you are not able to download a presentation, the publisher may have deleted the file from their server.
Safe-Lock to Baxter Connection
This inservice is not intended to replace the advice or training from your physician. Refer to the Instructions For Use (IFU) for detailed device description, instructions, contraindications, warnings, and precautions.
Fresenius Medical Care, DEL-CLAMP, Stay●Safe, Safe-Lock, DELFLEX and FREEDOM are trademarks of Fresenius Medical Care Holdings Inc. or its affiliated companies. All other trademarks are the property of their respective owners.
Baxter is a trademark of Baxter International Inc. or its affiliates.
Instructions on how to use Baxter products were reproduced from the manufacturer’s literature and were current at the time of print. Instructions are included for the patient's convenience but Fresenius Medical Care North America does not specifically endorse the use of these products or assume any liability related to their use. Should there be any differences between the instructions listed in this manual and the instructions for use provided with the medical devices referenced, the instructions provided with the medical device should be taken as accurate."
Note: for clarity the model is shown not wearing personal protective equipment (PPE). Follow your facility’s policies and procedures for use of personal protective equipment.
It is vital to maintain sterility of the connectors during connection procedures.
A Universal Connector will allow the patient with a Fresenius Safe-Lock PD system to use Baxter supplies while in the hospital.
Fresenius Safe-LockPD system
Please note Safe-Lock systems include:
Type II adapter on catheter
P/N # 050-30026
4 inch convertible extension set
Extension set with roller clamp
P/N # 050-30031
Premier transfer set
A Universal Connector (P/N #050-30026) or Pre-filled Universal Connector (P/N #050-30031) will connect to any of the patient catheter or catheter extension sets shown above.
Pre-filled Universal Connector
Baxter tubing solution set and related supplies for transfer set connection and exchange procedure.
Baxter extension set
4. Remove sealing cap
from large end of
5. Attach Universal
Connector to patient’s catheter
6. Remove the sterility protector
from the small end of the
7. Remove the sterility protector
from the Baxter extension set.
8. Attach the Baxter extension
set to the Universal
Connector. Proceed with
Typical connection and use of the Universal Connector
Choose the type of catheter or transfer set and the type of system utilized. Look down the column and across the row to the adapter needed. Click on the icon in the box to take you to the procedure
1-800-662-1237 if you have questions regarding the use of a Fresenius PD system or adapter.
800-323-5188 (P/N # 100676-01)