- 142 Views
- Uploaded on
- Presentation posted in: General

Principles of Fluorescence Spectroscopy

Download Policy: Content on the Website is provided to you AS IS for your information and personal use and may not be sold / licensed / shared on other websites without getting consent from its author.While downloading, if for some reason you are not able to download a presentation, the publisher may have deleted the file from their server.

- - - - - - - - - - - - - - - - - - - - - - - - - - E N D - - - - - - - - - - - - - - - - - - - - - - - - - -

XMUGXQ PFS0601

Principles of Fluorescence Spectroscopy

Chemistry Department

XMU

XMUGXQ PFS0601

Measurement of Fluorescence Lifetime

&

Time-domain Fluorescence

&

Frequency-domain Fluorescence

XMUGXQ PFS0601

7.1 introduction

7.2 pulse lifetime measurement

7.3 Phase and Modulation Measurements of Fluorescence Lifetime

7.4 Measurement of Time-resolved Decays of Fluorescence

7.5 Application of Time-resolved Fluorescence

7.6 Phase-sensitive Detection of Fluorescence

7.7 Application of PSDF

XMUGXQ PFS0601

Information given by fluorescence lifetime

- The frequency of collisional encounters

- The rate of energy transfer

- The rate of excited state reaction

- Information related to its environment

And so on

XMUGXQ PFS0601

I0

I0

t

t

- Pulse method

Light source

- Harmonic or phase-modulation method

Light source

XMUGXQ PFS0601

hvA

hvF

Log F(t)

or

log N(t)

S1

S1

knr

t

S0

XMUGXQ PFS0601

1 ns, 1

2 ns, 1

3 ns, 1

For a large number of fluorophores and small time interval, this sum becomes

XMUGXQ PFS0601

5 ns

50 ns

Pre-exponential factor

XMUGXQ PFS0601

Phase angle ()

Demodulation factor (m)

XMUGXQ PFS0601

=2f an important factor

For small lifetime, set large modulation frequency

For large lifetime, set small modulation frequency

Choosing modulation frequency, let

m = 0.3 ~ 0.7, = 30 ~ 70º

For commercial frequency-domain instrument, changing , measuring miand i, calculate average lifetime

XMUGXQ PFS0601

XMUGXQ PFS0601

Single component

multicomponent

Average lifetime

XMUGXQ PFS0601

I0

t

Pulse lifetime measurement

- Pulse width

Enough shorter compare to decay of fluorescence

ps, fs

- Photon counts

Enough for accurate measurement

Repeat excite

XMUGXQ PFS0601

Photomultiplier

Multichannel analyzer MAC

XMUGXQ PFS0601

Time to amplitude converter TAC

Multichannel pulse heigh analyzer MCPHA

XMUGXQ PFS0601

Simultaneous measurements of both wavelength and time resolved decays

XMUGXQ PFS0601

Log F(t)

or

log N(t)

t

In principle, for single exponential decay

XMUGXQ PFS0601

In practice, in consideration of the pulse width of lamp and multi-exponential decay

Intensity profile of light, L(t)

Intensity decay of fluorescence, F(t)

Measured intensity decay, R(t)

XMUGXQ PFS0601

In practicing measurement

- Measuring the lamp profile, L(t), by using a solution which scatters light

- Measuring the total intensity decay, R(t), by using the sample

At ti, a large number of pulses with equal width ti, each induce an impulse response in the sample

t - ti, emission delay compare to excitation

XMUGXQ PFS0601

Total intensity decay,

The purpose is to get F(t)

Commercial software available

XMUGXQ PFS0601

- Least –squares analysis of time-resolved decays

Let the number of components to be n,

Give initial values to i and i, and calculate, get

L(t), was measured

XMUGXQ PFS0601

The i and ivalues are varied until the best fit is observed.

In this expression the sum extends over the number (n) of channels or data points used for a particular analysis.

A minimum value of 2indicates the best fit.

XMUGXQ PFS0601

XMUGXQ PFS0601

Single exponential decay

Double exponential decay

XMUGXQ PFS0601

Single exponential decay

Double exponential decay

XMUGXQ PFS0601

I

t

- Deduct back ground

Measure intensity ex/em

Boxcar integrator

Target component

Sampling time

Back ground

Fast scan ex/ scan em

Light

Gated time

XMUGXQ PFS0601

荧光素标记荧光免疫分析，常常受到血液样品中胆红素背景荧光干扰，采用时间分辨荧光免疫分析可以有效地消除干扰。

荧光素寿命 3.6±0.46 ns, 测定时间 6.0 ns

胆红素寿命 0.21±0.14 ns, 测定时间 信号为零

XMUGXQ PFS0601

I

t

- Multi-components measurement

XMUGXQ PFS0601

[Q]

- Dynamic quenching

XMUGXQ PFS0601

- Time-resolved fluorescence immunoassay

Example

TBP-Eu3+

XL665

D

A

TBP-Eu3+

链[霉]亲和素

XMUGXQ PFS0601

Interference:

Background from media

Emission from free acceptor

XMUGXQ PFS0601

background

Free XL665

TBP-Eu3+(665nm/620nm)

FRET(665nm/620nm)

XMUGXQ PFS0601

XMUGXQ PFS0601

XMUGXQ PFS0601

XMUGXQ PFS0601

I(t)

mL=b/a

F(t)

mA=B/A

F(t)

mB=B’/A’

Principle

A<B

mL>mA>mB

A<B

XMUGXQ PFS0601

For single component

Excited with

Emission

Phase sensitive detection (lock-in amplifer )

D, detection phase

XMUGXQ PFS0601

F

F

At i

At i

F change with cos(D-)

F change with

XMUGXQ PFS0601

For two component

Emission

Phase sensitive detection

Phase depression

XMUGXQ PFS0601

XMUGXQ PFS0601

XMUGXQ PFS0601

XMUGXQ PFS0601

XMUGXQ PFS0601