# There are three major types of Blood flow Doppler patterns: 1) Continuous Wave 2) Pulse Wave 3) Color Flow - PowerPoint PPT Presentation

There are three major types of Blood flow Doppler patterns: 1) Continuous Wave 2) Pulse Wave 3) Color Flow

1 / 27
There are three major types of Blood flow Doppler patterns: 1) Continuous Wave 2) Pulse Wave 3) Color Flow

## There are three major types of Blood flow Doppler patterns: 1) Continuous Wave 2) Pulse Wave 3) Color Flow

- - - - - - - - - - - - - - - - - - - - - - - - - - - E N D - - - - - - - - - - - - - - - - - - - - - - - - - - -
##### Presentation Transcript

1. There are three major types of Blood flow Doppler patterns:1) Continuous Wave2) Pulse Wave3) Color Flow DOPPLER ULTRASOUND

2. What can We Measure with Doppler Ultrasound? • Velocities: Most calculations start with this parameter • Direction of Blood Flow • Pressure Gradients

3. The Doppler Principle is the basis of many applications

4. Beam alignment is crucial for accurate velocity measurements V = velocity of blood flow C = speed of sound in blood FT = transducer frequency FS = backscatter frequency Cos ϴ = cosine of angle between US beam & blood flow

5. As the angle of the incident beam and the direction of blood increases the measured velocity decreases and the error increases DOPPLER BEAM MEASURED BLOOD FLOW VELOCITY VECTOR WHEN BEAM IS NOT PARALELL BLOOD FLOW VELOCITY VECTOR

6. Effects of Intercept Angle in Velocity Measurements can not be ignored beyond 20 degrees

7. Pulse Wave, Continuous Wave & Color Flow

8. SPECIFIC APPLICATION OF DOPPLER MEASUREMENTS • Pulsed Wave Doppler • Stroke Volume Measurement • Continuous Wave Doppler • Peak velocity measurement • Color Doppler • Blood Volume flow & direction

9. Pulsed Wave Doppler • Measures the blood velocity at a specific location or sample volume • Exact localization of moving objects • Limited maximum velocity measurement because of Nyquist Limit (more on this later)

10. Pulsed Wave Doppler from transmittal flow to evaluate diastolic function

11. Continuous Wave Doppler • Measures the Maximum Velocity in a specific direction ( vs location for PW) • Essentially no limit in the maximum velocity measurement (no Nyquist limit) • The site of the maximum velocity cannot be localized

12. Continuous Wave Doppler MR JET VELOCITY PROFILE DURING SYSTOLE, NOTE DIRECTION AND MAX VELOCITY

13. Color Doppler • Multiple sampling with pulsed wave Doppler along the 2D scan lines. • The quantity and direction of velocity are measured in every point. • The intensity of the signal represents number of moving cells • Movement away from the transducer is blue and movement toward the transducer is red.

14. Color Doppler Parasternal long axis (LAX)

15. Blood Velocity Curves • Can be measured any where there is flow and an Acoustic Window • Basic calculus is used to analyze the velocity curves to obtain the Stroke Volume (SV) and Cardiac Output

16. PW vs. CW thru the Mitral Valve

17. Pulse Wave Thru Pulmonary Veins:Effects of Turbulence

18. CONTINUOUS DOPPLER WAVES CAN BE USED TO MEASURE:STROKE VOLUMECARDIAC OUTPUT

19. STOKE VOLUME WITH VELOCITY TIME INTERGRAL

20. STROKE VOLUME & CARDIAC OUTPUT INTERGRAL OF BLOOD VELOCITY DISTANCE TRAVELED BY BLOOD DISTANCE TRAVELED BY BLOOD AREA OF FLOW VOLUME OF BLOOD (STROKE VOLUME)

21. Stroke Volume Measurement thru LV out flow tract (LVOT) • SV = CSALVOT x VTILVOT SV = Stroke Volume CSA = Cross Sectional Area VTI = Velocity Time Integral LVOT = Left Ventricular Outflow Track