Download NowAn Image/Link below is provided (as is) to download presentation
Download Policy: Content on the Website is provided to you AS IS for your information and personal use and may not be sold / licensed / shared on other websites without getting consent from its author.While downloading, if for some reason you are not able to download a presentation, the publisher may have deleted the file from their server.
We finally managed to complement the beam tests of ULTIMATEwith the two sensors irradiated with 3*10^12 Neq/cm^2. One ofthem has a lso been exposed to 150 kRad 10 keV X-Rays. The chips were mounted on a beam telescope of ours, composedof 4 ULTIMATE sensors, and exposed to an electron beam at DESY(energy around 5 GeV). Data were collected with both sensorsmounted as DUT, at temperatures of 20~C ("normal conditions",abbreviated with NC on the plot) and 30~C. The measured detection efficiency and fake hit rate areshown on the attached figures for both chips, as function ofthe discriminator threshold setting (in mV). Also shown arethe values measured in Autumn 2011 at the CERN-SPS with twoother sensors, one non-irradiated and one irradiated with150 kRad 10 keV X-Rays.