Smart Flow Control Systems אפליקציות טיפוליות נוספות. System Contents. Catheter Set (for P49220 only) - SmartCatheter Peal away introducer Catheter Instructions. Compression Unit Compression Unit loaded over empty 500 ml bag Pump labels Medication labels .
Download Policy: Content on the Website is provided to you AS IS for your information and personal use and may not be sold / licensed / shared on other websites without getting consent from its author.While downloading, if for some reason you are not able to download a presentation, the publisher may have deleted the file from their server.
אפליקציות טיפוליות נוספות
Regulating Set(SmartInfuser Set ™)
Physician Guidelines and Instructions
SmartInfuser Set ™
The SmartCatheter and SmartInfuser Set ™ comprise the Administration Set.
Remove the SmartInfuserSet ™ and the SmartCatheter from the sterile packaging.
Connect the SmartCatheter to the Regulating Set using the luer connectors.
Tear off the bag port cover and insert the spike.
Verify the spike is positioned all the way in.
(3) Fill the Pump
Position the SmartReg® at C (Closed position).
Use aseptic technique to fill the pump.
After use, handle and discard in accordance with standard medical protocol.
(4) Prime the System
Position the SmartReg® at P (Priming position).
(5) Set the Regulating Set
Set the bolus volume by dialing the Push Button using the Key: the digit represents the particular ml selected for the bolus.
Lock the selected volume by pushing in the Safe Lock using the Key.
(7) Placing the Introducer
Care should be taken during catheter placement to assure that occlusion will not occur during use and that catheter removal will not be impeded.
Ensure that the catheter tip has not been placed in a vein or artery!
End of infusion tubing
Do not cover the filter of SmartInfuser Set™
Place the bolus over the wrist and secure it by tighten the strip.
Push the bolus button for an extra dose of medication when suffering exceptional pain.
Thank you for your attention