NSLS II Metrology R&D Activities. Peter Z. Takacs Experimental Facilities Advisory Committee Review 20 Oct 2006. Rationale for Metrology R&D for NSLS II. Every new advance in SR source design has driven improvements in optical components:
Download Policy: Content on the Website is provided to you AS IS for your information and personal use and may not be sold / licensed / shared on other websites without getting consent from its author.While downloading, if for some reason you are not able to download a presentation, the publisher may have deleted the file from their server.
Peter Z. Takacs
Experimental Facilities Advisory Committee Review
20 Oct 2006
Gratings, spheres, cylinders, paraboloids, ellipsoids, etc.
Add “real world” slope error to soft x-ray beamline KB mirror surface:
BNL LTP III measuring Si cylinder
QED SSI uses conventional Zygo Fizeau interferometer head combined with 6-axis positioning manipulator.
Measured and predicted 40nm image shape from SSI on 100mm long elliptical cylinder.
Yumoto,et al., RSI 76, 063708 (2005)
QED-developed algorithm solves for test surface error AND reference optics errors => self-calibrating, < 2nm residual errors.
View from interferometer
in stitching of Osaka elliptical cylinder
Elastic Emission Machining
Y. Mori, Osaka
High speed rotating tool