SRF Cryomodule 2 Cavity Test Results. Andy Hocker TD/SRF Dev Dept. All Experimenters Meeting, 10-OCT-2011. Context. ILC “S1” goal: 31.5 MV/m in an 8-cavity cryomodule DESY did it once with an XFEL prototype Yet to be demonstrated in Asia/Americas regions
Download Policy: Content on the Website is provided to you AS IS for your information and personal use and may not be sold / licensed / shared on other websites without getting consent from its author.While downloading, if for some reason you are not able to download a presentation, the publisher may have deleted the file from their server.
TD/SRF Dev Dept.
All Experimenters Meeting, 10-OCT-2011
ILC S0 spec
Q0 vs. Eacc
Field Emission (X-rays)
ILC S1 spec
Void in Cu plating
Dynamic compensation for Lorentz force detuning using fast piezoelectric tuners
CM2 under assembly at MP-9