Loading in 2 Seconds...
Loading in 2 Seconds...
Fresenius Medical Care Peritoneal Dialysis Connections. stay safe to Baxter Connection. Connections using Fresenius Medical Care products.
Download Policy: Content on the Website is provided to you AS IS for your information and personal use and may not be sold / licensed / shared on other websites without getting consent from its author.While downloading, if for some reason you are not able to download a presentation, the publisher may have deleted the file from their server.
staysafe to Baxter Connection
This inservice is not intended to replace the advice or training from your physician. Refer to the Instructions For Use (IFU) for detailed device description, instructions, contraindications, warnings, and precautions.
Fresenius Medical Care, DEL-CLAMP, Stay●Safe, Safe-Lock, DELFLEX and FREEDOM are trademarks of Fresenius Medical Care Holdings Inc. or its affiliated companies. All other trademarks are the property of their respective owners.
Baxter is a trademark of Baxter International Inc. or its affiliates.
Instructions on how to use Baxter products were reproduced from the manufacturer’s literature and were current at the time of print. Instructions are included for the patient's convenience but Fresenius Medical Care North America does not specifically endorse the use of these products or assume any liability related to their use. Should there be any differences between the instructions listed in this manual and the instructions for use provided with the medical devices referenced, the instructions provided with the medical device should be taken as accurate."
Note: for clarity the model is shown not wearing personal protective equipment (PPE). Follow your facility’s policies and procedures for use of personal protective equipment.
It is vital to maintain sterility of the connectors during connection procedures.
A staysafe Luer-Lock adapter, 4 inch will allow the patient with a Fresenius staysafe PD system to use Baxter supplies while in the hospital.
staysafe Luer-Lock adapter, 4 inch
Fresenius staysafePD system
staysafe Luer-Lock adapter, 4 inch (P/N #050-95003)
Baxter tubing solution set and related supplies for transfer set connection and exchange procedure.
Baxter extension set
Patient extension set
2. Remove the sterility
cap from the larger
end of the adapter.
3. Attach the adapter to the extension set.
4. While holding adapter, grasp the Baxter pull tab from the extension set sterility protector cap, pull to remove.
5. Remove the protector cap from the end of the staysafe luer-lock adapter.
6. Connect the Baxter extension set to the adapter. Proceed with the exchange following facility procedure.
1-800-662-1237 if you have questions regarding the use of a Fresenius PD system or adapter.
800-323-5188 (P/N # 100676-01)