Loading in 5 sec....

Chapter 1: IntroductionPowerPoint Presentation

Chapter 1: Introduction

- 210 Views
- Uploaded on
- Presentation posted in: General

Chapter 1: Introduction

Download Policy: Content on the Website is provided to you AS IS for your information and personal use and may not be sold / licensed / shared on other websites without getting consent from its author.While downloading, if for some reason you are not able to download a presentation, the publisher may have deleted the file from their server.

- - - - - - - - - - - - - - - - - - - - - - - - - - E N D - - - - - - - - - - - - - - - - - - - - - - - - - -

- Analytical Chemistry
- Qualitative analysis
- Quantitative analysis

- Classification of Analytical Methods
- Classical methods (wet chemical methods)
- Instrumental methods

- Types of Instrumental Methods
- Separation techniques
- Chromatographic
- Gas
- Liquid
- …….

- Electrophoresis
- Many types

- Chromatographic
- Detection techniques
- Optical spectroscopy
- Absorption
- Emission
- Fluorescence

- Mass spectroscopy
- Atomic
- Molecular

- Optical spectroscopy

- Separation techniques

- Instruments for Analysis

Signal

Generator

Sample

Analytical

Signal

Input

Transducer

(Detector)

Signal

Processor

Output

Transducer

(Readout)

- Instruments for Analysis

- Selection of An Instrumental Method
- Defining the problem
- Performance characteristics
Figure of Merits: Quantitative performance criteria of instruments that are expressed in numerical terms

- Precision
A measure of the random or indeterminate error

Relative Standard Deviation

- Bias
A measure of the systematic or determinate error

Bias = µ - Xt

µ - the population mean for the concentration of an analyte

Xt – true concentration

Standard reference materials (SRM)

Signal

Signal

x

x

x

x

x

x

x

x

Sbl

Sbl

Concentration

- Sensitivity
A measure of the ability of an analytical method to discriminate between small differences in analyte concentration

- Calibration sensitivity (S = mc + Sbl)

Concentration

Signal

Concentration

- Sensitivity
- Analytical sensitivity ( = m/sS)
m – slope; sS – standard deviation of the measurement

Relative insensitive to amplification factors

- Analytical sensitivity ( = m/sS)

Increase the gain of the instrument by a

Factor of two

Sm = Sbl + ksbl

Standard deviation of

the Blank signal

Analytical Signal

Blank signal

- Detection limit (Limit of detection, LOD)
The analyte concentration giving a signal equal the blank signal, Sbl, plus three time standards deviation of the blank, sbl

Convert signal response, Sm, to analyte concentration

Sm = Sbl + ksbl

Standard deviation of

the Blank signal

Analytical Signal

Blank signal

- Dynamic Range
Lowest concentration (LOQ) to the concentration where the calibration curve departure from linearity (limit of linearity, LOL)

Convert signal response, Sm, to analyte concentration

- Dynamic Range

- Guideline for Reporting Data (Recommended by ACS Committee of Environmental Improvement)
Analyte

ConcentrationRegion of reliability

< 3σRegion of Questionable detection

(unacceptable)

3σDetection limit

3σ-10σRegion of less certain quantitation

10σLimit of quantitation

> 10σRegion of quantitation

99.7%

Question: When the value of 10 sbl (standard deviation of the blank) is used as LOQ, what is the relative standard deviation of the measurement at this point at the 99.7% confidential level, and why?

- Selectivity
The degree to which the method is free from interference by other species contained in the sample matrix

S = mAcA + mBcB + mCcC + Sbl

The selectivitycoefficient for B with respect to A

kB,A = mB/mA

……………………………………

Note: this is the selectivity of an analytical detection technique

- Calibration Methods
Analytical response

- Comparison with Standard
- Direct comparison
- Colorimetric

- Titration

- Direct comparison

- Comparison with Standard

Analyte Concentration

Signal

Concentration

- Calibration Methods
- External Calibration Curve

Normally use the method of least squares

X

X

X

X

- Calibration Methods
- External Calibration Curve
Two Assumptions:

1. Linear relationship

2. deviation of the individual point from the straight line arises from the error in the measurement

- External Calibration Curve

Signal

Without sample matrix effect

With sample matrix effect (signal suppressed)

Concentration

- Calibration Methods
- Standard Addition

C1, C2, and C3 are the concentrations of analyte after spiked with stnard, with counting the amount of analyte in the original sample solution.

C0 is the concentrations of analyte in sample without spiking

- Standard Addition

C0

C1

C2

C3

Signal

x

S = mC + Sbl

Sample signal

Sbl

x

x

C0 =Sbl/m

x

Conc. of analyte C0

Concentration

- Internal Standard
Add a substance in a constant amount to all samples and, blanks, and calibration standard in an analysis

Sanalyte/Sinternal

Concentration