Syringe pump (LAMDA-P). What is the syringe pump?. A syringe pump is a small infusion or withdrawal pump, used to gradually administer small amounts of fluid for use in chemical and biomedical research . Specifications (Pump 11 Elite). Touch screen button.
Press the release button (1) and move the pusher block (2) to the approximate length of the syringe.
Unscrew the bracket clamping knobs (7,8) to loosen the retaining brackets (4,5) on the pusher block (2) and the Syringe holder block (3).
Unscrew the syringe barrel clamp by unscrewing the syringe barrel clamp knob (6).
Place the syringe on the syringe holder block (3). Ensure the barrel flange is with in the syringe holder’s block’s retaining bracket (5) and the plunger flange is in the pusher’ blocks retaining bracket.
Screw down the syringe clamp knob (6) until the syringe barrel clamp (10) is tight against the syringe barrel. Do not overtighten, especially when using glass syringes.
Secure the syringe flanges by pushing the retaining brackets (4,5) tightly against the syringe flanges using the bracket clamping knobs (7,8). The syringe plunger should be secured by the pusher block retaining bracket and the flange of the syringe should be secured by the syringe holder block retaining bracket.
Tighten the mechanical stop collars (9) using allenwrench (12) to prevent excess travel of the pusher block.