This presentation is the property of its rightful owner.
1 / 34

# 1979 Ventilator Mode Options PowerPoint PPT Presentation

Bourns BP200 (neonates) PC-IMV CPAP Bourns LS 104-150 (pediatrics) VC-CMV assist/control VC-IMV CPAP. 1979 Ventilator Mode Options. CMV CMV + ASSIST CMV + AUTOFLOW CMV + PRESS LIMIT SIMV SIMV + PRESS SUPPT SIMV + AUTOFLOW SIMV + PRESS SUPPT + AUTOFLOW + ATC PCV PCV + AUTOFLOW.

1979 Ventilator Mode Options

Download Policy: Content on the Website is provided to you AS IS for your information and personal use and may not be sold / licensed / shared on other websites without getting consent from its author.While downloading, if for some reason you are not able to download a presentation, the publisher may have deleted the file from their server.

- - - - - - - - - - - - - - - - - - - - - - - - - - E N D - - - - - - - - - - - - - - - - - - - - - - - - - -

#### Presentation Transcript

Bourns BP200 (neonates)

PC-IMV

CPAP

Bourns LS 104-150 (pediatrics)

VC-CMV assist/control

VC-IMV

CPAP

### 1979 Ventilator Mode Options

CMV

CMV + ASSIST

CMV + AUTOFLOW

CMV + PRESS LIMIT

SIMV

SIMV + PRESS SUPPT

SIMV + AUTOFLOW

SIMV + PRESS SUPPT + AUTOFLOW + ATC

PCV

PCV + AUTOFLOW

MMV

MMV + PRESS SUPPT

MMV + AUTOFLOW

APRV

CPAP

CPAP + PRESS SUPPT

### Key Ideas for Understanding Mechanical Ventilation

• Mathematical Models

• Equation of motion

• Time constant

• Mean airway pressure

• Control Variables

• Pressure, volume, dual

• Breath Types/Patterns

• Optimum Mode Selection

transairway

pressure

transrespiratory

pressure

transthoracic

pressure

volume

elastance = Dpressure / Dvolume

### Lung Mechanics

resistance = Dpressure / Dflow

flow

### Equation of Motion

ventilation

pressure

(to deliver

tidal volume)

elastic

pressure

(to inflate lungs

and chest wall)

resistive

pressure

(to make air flow

through the

airways)

=

+

Pmus + Pvent = Pelastic + Presistive

Pmus + Pvent = E x V + R x V

### Uses for the Equation of Motion

• Classify ventilators and modes

• ventilator controls only one variable at a time

• pressure, volume, or flow

• Monitor lung mechanics

• resistance & elastance (compliance), time constant

• proportional assist

• automatic tube compensation

Paw

Paw

Volume/Flow Control Pressure Control

Inspiration

Expiration

Inspiration

Expiration

20

20

Pressure

0

0

2

2

1

0

1

20

20

Volume

0

0

2

2

0

1

0

1

3

3

Flow

Time (s)

Time (s)

0

0

-3

-3

### Dual Control

• Use of both pressure and volume signals to control the breath

• Example: CMV+AutoFlow

• every breath is mandatory

• every breath is pressure limited, time cycled

• pressure limit is automatically adjusted to meet set tidal volume target using compliance measurements

• Contraindicated if large leaks

• Stable minute ventilation

• as with volume control

• Better patient synchrony

• as with pressure control

### Characteristics of a Mode

• Breathing Pattern

• Control variable

• Breath sequence

• Control Type

• Specific Control Strategy

• Phase variables

• Operational logic

### Breathing Pattern

• Control Variable

• Pressure

• Volume

• Dual

### Breathing Pattern

• Control Variable

• Pressure

• Volume

• Dual

• Breath Sequence

• Mandatory vs spontaneous

### Breath Types

• Spontaneous Breath

• Patient sets frequency and tidal volume

• Both patient triggered and patient cycled

• Mandatory Breath

• Machine sets frequency and/or tidal volume

• Machine triggered and/or machine cycled

### Breath Sequences

• Continuous Mandatory Ventilation

• CMV

• all breaths mandatory

• Intermittent Mandatory Ventilation

• IMV or SIMV

• mandatory and spontaneous breaths

• Continuous Spontaneous Ventilation

• CSV

• all breaths spontaneous

### Basic Breathing Patterns

Control VariableBreathing Sequence

Volume ControlContinuous Mandatory Ventilation

Intermittent Mandatory Ventilation

Pressure ControlContinuous Mandatory Ventilation

Intermittent Mandatory Ventilation

Continuous Spontaneous Ventilation

Dual ControlContinuous Mandatory Ventilation

Intermittent Mandatory Ventilation

Continuous Spontaneous Ventilation

### Potential Confusion

• All mandatory breaths are assisted

• Spontaneous breaths may be assisted or not

### Definition of “Assisted Breath”

• Assisted (ventilator does work on patient)

• Airway pressure rises above baseline during inspiration (or falls below baseline during expiration).

• Un-Assisted

• Airway pressure stays constant during inspiration or expiration.

• Loaded (patient does work on ventilator)

• Airway pressure falls below baseline during inspiration and rises above baseline during expiration.

### Assisted Spontaneous Breaths

• Pressure Support

• patient triggered, pressure limited, flow cycled

• patient controls timing and size of breath

• Automatic Tube Compensation

• patient triggered, pressure proportional to flow2, flow cycled

• active on inspiration and expiration

• patient controls timing and size of breath

### Characteristics of a Mode

• Breathing Pattern

• Control variable

• Breath sequence

• Control Type

• Setpoint, auto-setpoint, servo, adaptive, optimal

• Specific Control Strategy

• Phase variables

• Operational logic

### Control Types

• Setpoint

• user sets pressure, volume, flow, limit

• Auto-Setpoint

• ventilator chooses limit variable

• Servo

• ventilator delivers pressure in proportion to patient’s inspiratory flow

• ventilator chooses pressure setpoint to achieve operator selected volume setpoint

• Optimal

• ventilator chooses both volume and pressure setpoints

### Evolution of Ventilator Control Types

Tactical Control (within-breaths)

• setpoint (PC-IMV)

• auto-setpoint (Pmax)

• servo (Automatic Tube Compensation)

operator-selected, static setpoints

Strategic Control (between breaths)

• optimal (ASV)

ventilator-selected, dynamic setpoints

static model

Intelligent Control (between patients)

• knowledge based

• artificial neural network

ventilator-selected, dynamic setpoints

dynamic model

ability to learn from experience

CMV

CMV + ASSIST

CMV + AUTOFLOW

CMV + PRESS LIMIT

SIMV

SIMV + PRESS SUPPT

SIMV + AUTOFLOW

SIMV + PRESS SUPPT + AUTOFLOW + ATC

PCV

PCV + AUTOFLOW

MMV

MMV + PRESS SUPPT

MMV + AUTOFLOW

APRV

CPAP

CPAP + PRESS SUPPT

VOLUME

CONTROL

VC-CMV

VC-IMV

PRESSURE

CONTROL

• PC-CMV

• PC-IMV

• PC-CSV

DUAL

CONTROL

• DC-CMV

• DC-IMV

### Selecting the Best Mode (benefit/cost)

• Benefits

• Clinical

• maintenance of stable gas exchange

• reduced work of breathing

• optimum patient-machine synchrony

• Technical

• learnability (easy to understand)

• efficiency (minimum setup and maintenance)

• low error rate (inappropriate settings)

### Selecting the Best Mode (benefit/cost)

• Costs

• Clinical

• adverse reactions (lung stretch damage)

• associated complications (blood loss)

• Technical

• cost to patient (length of stay)

• cost to hospital (clinician interventions)

VOLUME

CONTROL

VC-CMV

VC-IMV

### Ventilator Mode Options

PRESSURE

CONTROL

• PC-CMV

• PC-IMV

• PC-CSV

DUAL

CONTROL

• DC-CMV

• DC-IMV

DC-CMV

DC-IMV

DC-IMV + PRESS SUPPT

DC-IMV + ATC (automatic tube compensation)

DC-IMV + MMV (mandatory minute ventilation)

DC-IMV + PRESS SUPPT + ATC

DC-IMV + PRESS SUPPT + MMV

DC-IMV + MMV + ATC

### Dual Control Options

Benefit

Cost

Mode

Clinical

Technical

Clinical

Technical

DC-CMV

-Stable tidal volume with each breath

-Least work of breathing because all breaths are equally supported

-Nearly optimum synchrony

-Easy to understand and monitor

-Fewest breath parameters to adjust during weaning or increasing support

-More stable intracranial pressure

-Requires fewer adjustments and perhaps fewer blood gases

Benefit

Cost

Mode

Clinical

Technical

Clinical

Technical

DC-IMV

-Stable volume for mandatory breaths only

-Increased WOB for spontaneous breaths

-Easy to understand and monitor

-Necessary to select correct frequency

-Less stable intracranial pressure

-Less stable minute ventilation

-May require more ABGs

-Weaning means “dumbing down” the ventilator

Benefit

Cost

Mode

Clinical

Technical

Clinical

Technical

DC-IMV + anything

-Stable mandatory breaths only

-Variable WOB for spontaneos breaths depending on options and settings

-Difficult to understand and monitor

-Necessary to correctly balance many settings

-Decreased operator efficiency

-Less stable intracranial pressure

-Less stable minute ventilation

-May require more ABGs

-Weaning means “dumbing down” the ventilator

DC-CMV

Tidal volume

infrequent

PEEP

FiO2

Weaning

automatic

### Ventilator Management Requirements

PC-IMV + Press Support

• Peak inspiratory pressure

• frequent

• PEEP

• Frequency

• Pressure support level (?)

• FiO2

• Weaning

• manual

• may be ineffective with PS

DC-CMV

Tidal volume

infrequent

PEEP

FiO2

Weaning

automatic

### Ventilator Management

DC-IMV + Press Support

• Tidal volume

• infrequent

• PEEP

• Frequency

• Pressure support level (?)

• FiO2

• Weaning

• mandatory breaths automatic

• spontaneous breaths manual

• more manual as rate decreases

### Thought For The Day

“In some ways we are as confused as ever, but we believe we are confused on a higher level and about more important things.”

A.R. Feinstein, MD

### WWW.AARC.ORG

e-learning system

at the AARC store