Introduction to 85108A Pulsed Network Analyzers. Pengcheng Jia Dec 13, 2001. Outline. What is measurement? What is 85108 Pulsed Network Analyzer? What are the Connectors & Cables? What are the Calibration Kits? What is Non-Insertable Calibration? How to do Pulsed Measurement?
Download Policy: Content on the Website is provided to you AS IS for your information and personal use and may not be sold / licensed / shared on other websites without getting consent from its author.While downloading, if for some reason you are not able to download a presentation, the publisher may have deleted the file from their server.
Dec 13, 2001
f, P, T, expected values and tolerance.
User setting, Gauging, Always use ESD Protection
LO source: 10M-20 G
RF source: 10M-50 G
Test Set: 1-50 GHz
Test Set works from 1-50 GHz in 2 Bands:
1-20 GHz Use directly LO source.
6-50 GHz Use third harmonic of LO source.
Step Mode: 85108A
Ramp Mode: 8722D
8722D with 3K IF BW
Never Exceed the Power Limitation on the Test Port
Port1 & Port2: +43 dBm
RF Input Port: +17 dBm
PSC: Precision Slotless Connector
(Disks are stored in Drawers, never remove them)
MD_1_20_PL, 1-20 GHz, Pulsed
MD_6_50_NP, 6-50 GHz, Non-Pulsed
MD_6_50_NP, 6-50 GHz, Pulsed
Swap Equal Between-Series Adaptors
Swap Equal In-Family Adaptors