Loading in 5 sec....

Special requirements for Photosources operating at PV electron scattering exp. PowerPoint Presentation

Special requirements for Photosources operating at PV electron scattering exp.

Download Presentation

Special requirements for Photosources operating at PV electron scattering exp.

Loading in 2 Seconds...

- 70 Views
- Uploaded on
- Presentation posted in: General

Special requirements for Photosources operating at PV electron scattering exp.

Download Policy: Content on the Website is provided to you AS IS for your information and personal use and may not be sold / licensed / shared on other websites without getting consent from its author.While downloading, if for some reason you are not able to download a presentation, the publisher may have deleted the file from their server.

- - - - - - - - - - - - - - - - - - - - - - - - - - E N D - - - - - - - - - - - - - - - - - - - - - - - - - -

International Workshop

PAVI 2006

Milos Island

20/05/2006

by

Kurt Aulenbacher

Institut für Kernphysik

der Uni Mainz

B2/A4 collaboration

- The problem
- HC-intensity asymmetry
- Sources of other HC-fluctuations
- Low energy polarimetry

Necessary, but not specific for PV-

Experiment.

1.) Reliable beam production at desired intensity level

2.) Provide desired spin orientation

3.) High Polarization (>80%)

I.) Polarization meas. DA/A ~DP small

limiting factor in several PV-exp.

II.) HC-control A always important

limiting especially when A<10-6

Source team can provide support for point (I),

(II) is more important.

Scattering experiments (simplified)

T

S

A

- measure P accurately! (I)

D measures R+-

q

+

Let x be a vector formed from the relevant parameters:

{

:=HCA

Goal: Error of HCA should be small against other error contributions. (DP, Dstat)

1.) The (average) values of xi+-xi- have to be measured with good accuracy.

good stability of xi + high spin flip frequency desirable

2.) Relative sensitivities have to be determined and are only known with limited accuracy.

Higher order coefficients usually not well known.

(xi+-xi-) should be “small” (i.e. sufficiently close to zero).

HC-Control schematics (PVA4)

(Almost) no active HC-compensation, except by stabilization!

Adjust to zero crossing &

Observe stability!

Sketch of polarization optics

Result measuring I-HCA=(I+-I-)/(I++I-)

Assuming analysing power of

Photocathode, imperfections in the alignment and in the phase shifts (birefringences) of the optical

Elements (similar to Humensky et al. NIM A 521 (2004) 261)

Description with 4X4 polarization transfer matrices:

For ‘thin’ cathodes: I+- ~ S+-0

Expand Matrix elements to first order in the imperfections:

Predicted I-HCA as function of compensator rotation angle b

AISR=Analysing power of cathode, with polarizer axis oriented at 2qk,

Measured for several high P cathodes: AISR=0.02-0.05

fa= f++f-/p: Normalized asymmetric phase shift of pockels cell

(forced zero crossing!),fa=0.03 (typ.)

e3=circular stokes component of light at input of Pockels cell,

Not measured, est. to <0.003

s~0.999 = diagonal polarisation component at input of Pockels cell,

fc= deviation of half wave plate from 180 degree retardation

0.01 (quote by company)

D,c=function of birefringence of optical elements between PC and cathode.

(measurable, D~0.01)

1.) Stability does not depend on the symmetric phase shift error (f+-f-)-p

2.) Parameters extracted from fit in agreement with reasonable values

of optical imperfections

3.) Introducing an additional half wave plate

(General sign changer) will also change I-HCA.

1.) Absolute value of phase shift does not contribute to IHCA (in first order)

2.) Asymmetric phase shift + compensator temperature dependence!

3.) Sensitivity depends on steepness of zero crossing

4.) Reduction of sensitivity due to stabilization! (1/G~2-10)

From fit-curve:

Realistic only if second order effects (HC-Transmission changes) do not occur

{

Offset/(4b amplitude) while varying qk:

Offers to reduce

Problem by order(s)

of magnitude….But

AISR is the analyzing power of the photocathode which will depend on the photocathode

type (composition, thickness,,,) typically: superlattice 2%, strained layers 4%, GaAs:<0.2%.

1.) Why did PV-experiments before 1990 observe large

asymmetries and position fluctuations

with very small analyzing power of the photocathode?

2.) What is the origin of the HC-fluctuation

of other parameters like position, angle, energy?

Lock-

in

Pulser

Sw.

He/Ne Laser

Detektor with

low analysing power

PC

Experiment results in I-HCA of 10000 ppm

(no lock in needed!)

Luck! The signal is so large that it´s easy to find a reason….

Screen

Prism

Backreflexions for the different helicity states.

- For scattering centers at different positions the ability to interfere (at an image point) is changed by switching the helicity.
- The interference pattern on the photocathode is therefore also helicity dependent, especcially in the ‘halo’ of the laser beam

Lock-

in

Pulser

Sw.

diode-laser

Movable Detektor

with pinhole

PC

- Helicity correlated movement of centroid is 1mm.

LOW-E polarimetry provides some support for the experiment if it can be done convienently and fast!

- Goal 1: fast relative measurement at full current with good reproducibility
- Goal 2: accuracy < 2%

Measurement time < 2min @1% stat. Acc. @20 mA

Beam installation time req: (40min) will be reduced to <15min.

Asymmetry vs. Spin rotator angle (164 Grad)

8 hour measurement of asymmetry

Theo:

Low energy: Fink et al.: Phys Rev A (38,12), 6055 (1988)

‚High‘ energy: Uginicius et al.:Nucl Phys A 158 418 (1970)

Exp:

Low energy: Gray et al.: Rev. Sci. Instrum. 55,88 (1984)

High energy: Sromicki et al. Phys. Rev. Lett. 82,1, 57 (1999)

Z=79

Analyzing power can be calculated with less than 1% accuracy

Energy variation

at fixed scattering

angle

m

- Low energy may be very accurate (DP/P < 1%)
(Mayer et al. Rev.Sci. Inst. (64,952(1993))

- Always possible to achieve low set-up time
- Spin losses under control <<1%
- Spin orientation can be calculated to <1 deg.
- Measurement at full exp.current possible and fast.
- Calibration check may be handled as accelerator ‚service‘ good calibration tracking.

- HC-effects do contribute to, but do not dominate
- the error budget (at PVA4).
- 2. Stable operating conditions have to be achieved, if necessary
- extensive stabilization systems have to be used
- light optical effects are rather complicated but ‘treatable’
- Better understanding + technology offers potential
- to keep situation acceptable also for future exp.