PULSION Medical Systems AG. PiCCO 2 vs. Edwards EV1000. Edwards EV1000 – Setup. VolumeView sensor VolumeView femoral arterial catheter VolumeView thermistor manifold CVC standard TruWave pressure transducer EV1000 clinical platform. Patient Monitor. AP. CVP.
PiCCO2 vs. Edwards EV1000
Additional transducer for injection detection is neccessary!
EV1000 basic setting is more complex than a PiCCO basic setting!
Instructions for use
VolumeView Catheter Set
VolumeView Transducer Set
CVP Transducer + Injectate Sensor
“Fancy” guidewire dispensor
*tested with PiCCO2, Philips Intellivue, Dräger Smartpod
ELWIsmart compensation after lobectomy Clinical studies necessary to prove validity and accuracy, until today three studies on this topic showing no significant EVLW differences before and after lobectomy
SVVextra Already in the PiCCO algorithm since PiCCOclassic V4.1 (from 2000); means, the Edwards SVV algorithm was and is still erroneous
Compensation of arrhythmias in AP Curve
4. “Correct” calculation
GEDVsmart, EVLWsmart EV1000 EVLW validation was done by using PiCCO as gold standard with excellent agreement, means no significant improvement in the new algorithm.
GEDVsmart should enhance accuracy in case of a L-R shunt. No data published regarding this.
Right-to-left heart shunt detection also possible with the TD curve in the PiCCO, until today only one publication (letter by Michard), no quantification of shunt volume or cardiac output, which was already included in PiCCOplus V7.1 but skipped in PiCCO2 because of not proven clinical relevance and accuracy
Module available (soon)
Parameter connection available