- 54 Views
- Uploaded on
- Presentation posted in: General

Electron detectors for HAPPEX2 and He

Download Policy: Content on the Website is provided to you AS IS for your information and personal use and may not be sold / licensed / shared on other websites without getting consent from its author.While downloading, if for some reason you are not able to download a presentation, the publisher may have deleted the file from their server.

- - - - - - - - - - - - - - - - - - - - - - - - - - E N D - - - - - - - - - - - - - - - - - - - - - - - - - -

4

- Design
- Tests with high E electrons
- PMT tests
- Installation in HRS

t = 65 Mhz

Dose = 2 MRad

100 x 600 mm

Detector stack overlaps elastic line

Two segments geometry for

HAPPEX2

5 inches PMT

Brass-quartz stack

Light guide

Filter box

- Radiation hard (Babar, CMS, E158)
- Optical polished
- (15-20 Angstr)
- 92% transmission in the range
- [350,550] tested with optical
- bench and with cosmic rays

Bext

False asymmetry

Fe

Fe

N+

N-

FA= M(Fe) x P(Fe) x <A>

Polarized electrons

M(Fe) -> specified ~ 1%

P(Fe) -> a) Bext at focal plane

b) mr

<A> Mean Moeller asymmetry -> numerical calculations.

- -Bext <= 0.4 Gauss, mr <=2
- PFe < 10-5
-<AMoller> << 1 ppm

- -Negligible contribution in
- the detector asymmetry.

Moeller dipole

Beam pipe

h2

h1

Motorized rail

Trigger scintillators

Hodoscope

Selection of 2 e- of same energy

- Nge/e- = 250

DE/E = 18 %

100 x600 mm

DE/E=20%

- Larger resolution balanced
- by extra statistics.
- Need to correct for
- correlations between two
- segments for final result.

Real data to

be analyzed

DE/E=90%

- PMT burle 8854
- Need wide range in gain
- (10E3-106) :
- 2 bases available:
- - Happex
- - Resistive, low gain

Gain/IAnode correlations

Running conditions:

Preliminary results:

40 MHz, Filter 1/10

25 ge/e

I=1mA with G=104

-Happex Base DG/G =5% @ 1170V

-Resistive base: DG/G= 2%@ 1420V

Horizontal rails

10 cm vertical gap between

VDC and detector for scanners

- Remove 1 segment in each spectro. ~4h
- Start installation of previously prepared “L” detector.
- Assembly of the 2 removed segments for the 2nd “L” ~8h
- Installation of the 2nd “L”~6h
- Could be done within 1 day
Man power : 5

Eveything at:

www-dapnia.cea.fr/Phys/Sphn/Parity/