「光検出器 MPPC の 応答曲線の大量測定 A large amount of measurement of response curve of photodetector MPPC. University of tsukuba Hiroko koike 筑波大学 6/18 小池 博子. 1. y ： ADC pulse height x ： rotation number of polarizing plate （ × ４．５ ° ）. motivation. MPPC. PMT.
Download Policy: Content on the Website is provided to you AS IS for your information and personal use and may not be sold / licensed / shared on other websites without getting consent from its author.While downloading, if for some reason you are not able to download a presentation, the publisher may have deleted the file from their server.
「光検出器MPPCの応答曲線の大量測定 A large amount of measurement of response curve of photodetector MPPC
University of tsukuba
y： ADCpulse height
x：rotation number of
Shape is sure to fall into disorder if mppc is a saturated condition.
The measuring method of the response curve is established.
A flat area is seen in mppc plot. The saturated condition was able to be confirmed.
by the same sample.
1, Reproducibility of polarizing plate and laser.
2, Reproducibility when mppc is pulled out and opened.