TS-1000 ELECTRONIC TORQUE METER. State-of-the-art torque meter for: Service and Repair Production and Assembly Torque Wrench Calibration. Click here to run demo. Click here for safety warnings. HOW IT WORKS.
Click here to run demo
Click here for safety warnings
The electronic torque meter connects to different types of torque transducers and displays the torque in real time torque units on a large digital LCD.
You can read the actual applied torque with great accuracy.
You can quickly set a target torque so the meter will prompt you if thetorque is within the desired limits.
One type of transducer, the in-line torque transducer, attaches between a wrench and the socket just like a conventional extension.It can be installed on a torque wrench to verify the output or record the torque
or it can be installed on a conventional ratchet and used as an accurate digital torque wrench.
Use straight female-male adapters if needed with no change in accuracy
Perfect for Calibration Labs.
All transducers are “plug-and-play”
Auto range detect
Auto calibration set-up
Auto Units set-upMIX AND MATCH TRANSDUCERTYPES WITH ONE METER
You can expand the system as your needs change
1. Turn meter off
2. Plug in transducer
3. Turn meter on
4. Apply torque with transducer
Clears torque on the display
Saves torque on display
The display will look like this when the meter goes into the battery saving-power down mode.
Press the “clear” key to wake the meter up
There are three display modes to choose from:
Each transducer can read in the corresponding unitsso there is no need to buy different Metric or American transducers.
The limit values can be set as a % of a target torquefor example, 4% of 100 ft.lb.
Or they can be set as actual high and low values for example, 96 ft.lb. to 104 ft.lb.
The type of tolerance used depends on how you set up the meter under Menu (default is % of the target torque)
Tolerance is set to 4%
Target torque is 100 ft.lb.
Press the “limits” key once
To turn on the limits function
RED Light turns on when torque is over 104 ft.lb.
GREEN light turns on when torque is between 96 and 104 ft.lb.
YELLOW light turns on when torque is under 96 ft.lb.
Press and hold the “limits” key for 1 second to set a new target torque.
See next slide for instructions
After pressing and holding the “limits” key the current target torque will show. You can now quickly set the new target torque value.
Here is how you set the value to 100.0
Make a habit of zeroing the display, even when the display shows “zero”
Make sure there is no torque on the transducer when pressing the zero/tare keys.
Press “clear” key when finished viewing
Range (50 in.lb.)
Cal factor (millivolts/volt)
There are 7 more advanced features under ‘Set-up Menu.’
Press “mode” and “units”to enter menu set-up
Press “save” to select shown set-up option
Set Tolerance type (% or Hi/lo)
Press “clear” to exit
Auto Reset method
The display will look like this
Press “next” key to scroll throughset-up functions
This function recalls the values in memory for you to view
After pressing “save” from menu location ‘Recall’ you will see this on display
This is the current number of values stored in memory
Press “next” key toscroll through memory
Press “clear” when finished
This mode sends all values in memory out the RS232 serial port for printing or sending to a computer.
As the data is sent the data number will count on the LCD
See operation manual and Tech Tips for information on how to setup your printer or computer.
This function erases all torque data saved in memory
From Set-up press “save” when Erase is displayed
Press “save” again to erase
Press “clear” to abort erase and return to operation
(See the manual for a more detailed description on filters).
Press “next” to select a new filter
Press “save” to save the filter setting
This function will pre-set the default tolerance mode
You can choose between % of target or High and Low
Refer to the operation manual for details on how to set the default tolerance mode
This option sets up how you want the LCD to clear the torque value on the display during operation
Saves new Reset method
See operation manual for a full description of reset methods
If you select this mode accidentally, turn the meter off, then on again to get out of calibration mode without changing calibration factors.
That is it! You can see that this torque meter is very powerful, easy to learn and use.