1 / 11

# VOLTMETER. Peukur tegangan dipasang paralel Rm  besar ( tahanan idaman ~ ) - PowerPoint PPT Presentation

VOLTMETER. Peukur tegangan dipasang paralel Rm  besar ( tahanan idaman ~ ) Arus mengalir lewat peukur sangat kecil. 10 k Ω. 10 k Ω. 5 k Ω. V. 5 k Ω. Rm = 3 k Ω. 10 V. 10 V. V ’. V. V BU = 10 V. Va-b = (5 / 10+5 )x 10 V = 3,33 V Va-b (terukur Voltmeter dgn Rm = 3 k Ω )

I am the owner, or an agent authorized to act on behalf of the owner, of the copyrighted work described.

## PowerPoint Slideshow about ' VOLTMETER. Peukur tegangan dipasang paralel Rm  besar ( tahanan idaman ~ )' - aristotle-olson

Download Policy: Content on the Website is provided to you AS IS for your information and personal use and may not be sold / licensed / shared on other websites without getting consent from its author.While downloading, if for some reason you are not able to download a presentation, the publisher may have deleted the file from their server.

- - - - - - - - - - - - - - - - - - - - - - - - - - E N D - - - - - - - - - - - - - - - - - - - - - - - - - -
Presentation Transcript

• Peukur tegangan dipasang paralel

• Rm  besar ( tahanan idaman ~ )

• Arus mengalir lewat peukur sangat kecil

10 kΩ

10 kΩ

5 kΩ

V

5 kΩ

Rm = 3 k Ω

10 V

10 V

V’

V

VBU= 10 V

Va-b (terukur Voltmeter dgn Rm = 3 kΩ )

Rin kemudian : Rm // R2=(3x5)/8 = 15/8 = 1,875 kΩ.

Va-b = 1,875 / (1,875 + 10) x10 V = 1,57 V

Teg sesungguhnya 3,33 V, terbaca : 1,57 V.

tegangan sesungguhnya.

Dengan menambah sebuah atau beberapa tahanan secara seri dengan instrument dapat merobah batas ukur instrument tsb.

V = Im (Rs + Rm)  Rs = (V-Im.Rm) / Im

Rs = (V/Im ) - Rm

Rs

I

Im

Gerak

D’ Arsonval

Rm

R1

Im

V2

R2

Multi range Voltmeter:

R1= (V1/Im) – Rm

R2= (V2/Im) – Rm

R3= (V3/Im) – Rm.

R1= (V1/Im) – Rm

R2= (V2/Im) – (Rm+R1)

R3=(V3/Im)–(Rm+R1+ R2)

V3

R3

R3

R2

R1

V3

Im

V2

V1

Rm

I

PENGARUH PEROBAHAN TEMPERATUR

Tahanan seri Rs tidak ter pengaruh perobahan suhu.

Rm terpengaruh dengan koefisien α pada (t2– t1)

I = E / [Rm{1+α(t2 – t1)} + Rs]

= E / (Rm + Rs) x { 1 – (α.Rm)/(Rm+Rs) x (t2– t1)

Perubahan arus yang terjadi akibat perubahan temperatur:

(α.Rm) / (Rm+Rs) x (t2 – t1) x 100%

E

Rm

Voltmeter dengan Rs = 45 Ω Rm= 5 Ω masing2 terbuat dari bahan manganin dan tembaga. Koefisien temperatur Rs = 0 %/oC dan αuntuk Rm = 0,4 %/oC pada perobahan dari 20 oC ke 30 oC

Penyelesaian:

(α.Rm) / (Rm+Rs) x (t2 – t1) x 100%

(0,004x5) / (45+5) x10 x100% = 0,4 %

Terjadi kesalahan baca akibat perobahan arus

mengalir sebesar 0,4 %

Input resistan dari voltmeter yang paling simpel adalah: Rin = Rs + Rm atau

Rin = V / Im

Faktor sensitivitas =>

S di defenisikan 1/arus skala penuh.

S = 1/Im atau S = 1/ A

= 1 / V/ Ω = Ω / V

Rm = S x V

R1

V

R2

Va-b = (200 / 200 + 200)x 100 V = 50 V

Va-b (terukur Voltmeter dgn S = 20 kΩ/V) :

mempunyai => Rin = S x V = 20 kΩ/V x 50 V => 1000 kΩ

kemudian : Rin // R2=(200x1000)/1200 = 200.000 / 1.200

= 166,667 kΩ.

Va-b = 166,667/(200 + 166,667) x`100 V = 45,45 V

Teg sesungguhnya 50 V, terbaca : 45,45 V.

tegangan sesungguhnya.

100 V

200 kΩ

S=20 k Ω / V

Rin =

Rs+Rm

Vo

AKURASI:

Kv = Vw / Vo ;

I = Vo / (Ro + Rin)

Vw = teg terukur

Vo = teg sesungguhnya

Tegangan terukur :

Vw = I. Rin atau Vw = {Rin/(Ro+Rin)} x Vo

Jadi Kv = Vw / Vo

Kv = Rin / (Ro + Rin)

I

V

Vw

V

R2

Contoh:

Vo = (200 / 200 + 200)x 100 V = 50 V

Ro ( Requvalenbila dilihat dari a-b) =>

Ro = (200x200)/(200+200) = 100 kΩ

Rin = S . V = 20 kΩ/V x 50 V => 1000 kΩ

Vw = Rin / (Ro + Rin) x Vo =>

Vw = 103 / (100+ 103) x 50V =>Vw = 45,45 Volt

Kv = Vw/Vo = 45,45/50 x 100%

Kv = 91 %.

100 V

200 kΩ

S=20 k Ω / V