This presentation is the property of its rightful owner.
1 / 21

# HBD Transmission Monitor Update III: Noise Analysis PowerPoint PPT Presentation

HBD Transmission Monitor Update III: Noise Analysis. HBD Working Group Meeting 10/10/06. B. Azmoun & S. Stoll BNL. Low Sensitivity at either end of Trans. plot. Low Sensitivity in regions of spectrum where light output is low In these regions the Trans. Calc is more susceptible to

HBD Transmission Monitor Update III: Noise Analysis

Download Policy: Content on the Website is provided to you AS IS for your information and personal use and may not be sold / licensed / shared on other websites without getting consent from its author.While downloading, if for some reason you are not able to download a presentation, the publisher may have deleted the file from their server.

- - - - - - - - - - - - - - - - - - - - - - - - - - E N D - - - - - - - - - - - - - - - - - - - - - - - - - -

## HBD Transmission Monitor Update III: Noise Analysis

HBD Working Group Meeting 10/10/06

B. Azmoun & S. Stoll

BNL

### Low Sensitivity at either end of Trans. plot

• Low Sensitivity in regions of spectrum

• where light output is low

• In these regions the Trans. Calc is more

• susceptible to

• Incorrect Subtraction of the DC

• Fluctuations in the Bckgnd “Noise”

• Component

• Signal consists of 3 Components:

• Background Noise

• Dark Current (DC)

• pe Signal

B. Azmoun

### How Trans. Is Calculated as a function of Wavelength

Expression for Transmittance Calculation:

S: Sample Scan Current (Gas)

R: Reference Scan Current (Vac)

Ms: Monitor Current during sample scan

MR: Monitor Current during ref. scan

D: Corresponding Dark Current (Const.)

• The PMT current at each wavelength is measured during an automated wavelength scan, where the scan progresses in 1 nm steps over a range from 110 to 190nm.

• The current value at each wavelength is taken as the mean of 100 ammeter readings, corresponding to a sampling time of 17 sec. (including digital and analog filtering)

• The PMT DC is NOT measured point-by-point, but only for 2 data points before and after the scan (as you’ll see later, this has to be improved). The four data points are averaged and this constant is then subtracted off of the current value at each wavelength: IPMT(l) – DC.

• As specified above, the transmittance calc. is then a ratio of the pe current (with the DC subtracted off) measured when the cell is filled with gas and under vacuum. This ratio is then normalized to a ratio of lamp monitor measurements, each taken simultaneously with the corresponding Sample or Ref. measurement.

B. Azmoun

### Effect of DC on Trans. Calc.

• The Trans. Calc. becomes strongly dependent on an accurate

• measurement of the DC once the DC value becomes comparable to the pe signal

• Easy Fix: measure DC for more than just 4 data points perhaps 20?

Ref. DC Underestimated

• The plots to the left represent the same measurement with different (artificial) values of DC subtracted off the pe signal

• Depending on which of the four DC values in the Trans. expression are measured incorrectly, the Trans. plot will exhibit either an effective absorbance, or an effective Trans. greater than 100%

• Even once the DC is subtracted off correctly, small systematic variations in the Trans. may remain, and represent the small (yet relatively significant), and slow variations in the DC [These systematic variations reveal the limits of the approx. made in the Trans. Calc., namely the assumption that the DC is constant—here it pays to have a PMT w/ low DC]

Ref. DC Overestimated

Ref. DC  ~ Flat Trans.

B. Azmoun

## Scans of Noise, Dark Current and pe Current for each PMT of each Cell of Trans. Monitor

• Similar to the systematic modulation of the Trans. curve caused by the incorrect subtraction of a constant, fluctuations in the PMT current will also have a strong impact on the Trans. curve in regions of low light. Once the fluctuations in the currents become comparable to the pe signal + DC, the change to the transmittance may be represented by the following expression, which includes a time dependent component, d(t).

As is shown in the following plots, this time dependent component has a common source: External noise coupling to the DC + pe signal. Therefore, dS = dR = dMS = dMR.

CELL 1

### Mon. PMT = PA0085Cell PMT = PA0089

B. Azmoun

Noise & DC Spectra

B. Azmoun

CELL 2

Mon. PMT = PA0085Cell PMT = PA0088

B. Azmoun

Noise & DC Spectra

B. Azmoun

CELL 3

Mon. PMT = PA0085Cell PMT = PA0086

B. Azmoun

Noise & DC Spectra

B. Azmoun

Lab spectrometer

Mon. PMT = PA0038Cell PMT = PA0080

B. Azmoun

Noise & DC Spectra

B. Azmoun

~45min

B. Azmoun

### Summary of Observations

• The Noise of both the “Monitor” and “Sample” are well correlated

• The sigma for both (Noise) and (Noise + DC) are strongly correlated.

• The Trans. Mon. DC is ~const.

•  sigma(Noise + DC) – sigma(Noise) ~ ZERO

• The mean DC of (at least one of) the Lab PMT’s is about the same as the Trans. Mon. PMT’s (~10pA). BUT, the fluctuations of (DC + Noise) for the Lab Spect. are a factor of ~100 less than for the Trans Mon. ( tens of pA Vs tenths of pA), due to the fluctuations in the noise component of (Noise + Signal), not due to the DC component.

• Thus, the Trans. Monitor PMT’s are probably Ok.

B. Azmoun

Sigma/Mean: Comparison of Lab

Spectrometer to HBD Trans Mon.

• Sigma/Mean of HBD Trans Monitor is similar to that of the Lab spectrometer, even

• though the Lab spect. has about a factor of 10 - 20 less light…the larger fluctuations

• in the HBD Trans. Monitor offset the increase in light output.

B. Azmoun

Long Duration Scans (DC & Signal)

B. Azmoun

Sigma

B. Azmoun

Noise & DC Spectra Vs Time

B. Azmoun

### Mean DC and Avg. Fluctuation Vs Time

Even over a very long duration (~13hrs.) DC is ~Constant, while

the fluctuations in the Monitor and Sample Scans remain

Strongly Correlated

B. Azmoun

### To Do …

• Investigate Noise Issues

• Lab Noise is a problem in general

• What is different between Lab Spect. and HBD Trans monitor

• Flow CF4 and check compatibility with Al collimation tube (coated with black layer of ???)

B. Azmoun