TROBLESHOOTING GAS CHROMATOGRAPHY (GC). Parto Zist Behboud Reza Daj 8/16/2014. Today’s Overview. Introduction. Injection Port. Flow Controller. Detector. Data Processor. Electric Signal. Column Oven. Gas Cylinder. Column. Troubleshooting Approach. Sample Injector System.
Download Policy: Content on the Website is provided to you AS IS for your information and personal use and may not be sold / licensed / shared on other websites without getting consent from its author.While downloading, if for some reason you are not able to download a presentation, the publisher may have deleted the file from their server.
TROBLESHOOTING GAS CHROMATOGRAPHY (GC)
Parto Zist Behboud
Carrier Gas and Detector Gas
He and N2 are usuallyused.
(In the case of a capillary column, helium is desirable)
The gas of 99.999% or more of purity is used.
TCD ... No need
FID ... H2, Air
ECD ... N2(In the case of a packed column, carrier gas is surely N2.
In the case of a capillary column, N2 is used for makeup gas. )
FTD ... H2, Air
FPD ... H2, Air
BID ... He
Drifting or Cycling Baseline
Contact: [email protected]