TROBLESHOOTING GAS CHROMATOGRAPHY (GC). Parto Zist Behboud Reza Daj 8/16/2014. Today’s Overview. Introduction. Injection Port. Flow Controller. Detector. Data Processor. Electric Signal. Column Oven. Gas Cylinder. Column. Troubleshooting Approach. Sample Injector System.
Download Policy: Content on the Website is provided to you AS IS for your information and personal use and may not be sold / licensed / shared on other websites without getting consent from its author.While downloading, if for some reason you are not able to download a presentation, the publisher may have deleted the file from their server.
Parto Zist Behboud
Carrier Gas and Detector Gas
He and N2 are usuallyused.
(In the case of a capillary column, helium is desirable)
The gas of 99.999% or more of purity is used.
TCD ... No need
FID ... H2, Air
ECD ... N2(In the case of a packed column, carrier gas is surely N2.
In the case of a capillary column, N2 is used for makeup gas. )
FTD ... H2, Air
FPD ... H2, Air
BID ... He
Drifting or Cycling Baseline