Loading in 5 sec....

Horizontal Distance MeasurementPowerPoint Presentation

Horizontal Distance Measurement

- 668 Views
- Updated On :
- Presentation posted in: General

Horizontal Distance Measurement. Tape Odometer Subtense Bar Stadia EDM. Distance Measurement. Devices and accuracy:. Older technologies “ quick look” Pacing : 1:50 Optical rangefinders: 1: 50 Odometers: 1:200 Tacheometry / stadia 1: 500 Subtense bar 1: 3000 Tapes: 1: 10,000.

Horizontal Distance Measurement

Download Policy: Content on the Website is provided to you AS IS for your information and personal use and may not be sold / licensed / shared on other websites without getting consent from its author.While downloading, if for some reason you are not able to download a presentation, the publisher may have deleted the file from their server.

- - - - - - - - - - - - - - - - - - - - - - - - - - E N D - - - - - - - - - - - - - - - - - - - - - - - - - -

Horizontal Distance Measurement

Tape

Odometer

Subtense Bar

Stadia

EDM

Devices and accuracy:

- Older technologies “ quick look”
- Pacing : 1:50
- Optical rangefinders: 1: 50
- Odometers: 1:200
- Tacheometry / stadia 1: 500
- Subtense bar 1: 3000
- Tapes: 1: 10,000

- Modern Technology:
- - Electronic Distance Measuring (EDM) devices: 2 +2 ppm

- Accuracy and speed considerations for civil engineers.
- Sources of Errors:
- Incorrect length of the tape
- Temperature difference
- Sag
- Poor alignment
- Tape not horizontal
- Improper Plumbing

- The idea of an odometer.
- Subtense bar: a 2 m rod.
- Distance H= cot(/2) m.

Aiming Telescope

- Subtense Bar
- Distance H = cot(/2) m.

L

L /2

H

/2

tan ( /2) = (L/2) / H

H = (L/2) / tan( /2)

If L = 2 m, then

H = 1 / tan (/2) = cot (/2)

Example:

what is the horizontal distance between A and B if the angle was 2?

- Chapter (16)
- Principle of the Stadia:
- Horizontal Distance = 100 rod interceptfor a horizontal line of sight and a vertical rod
- Symbols:
- (I) rod intercept, or stadia interval
- (i) spacing between stadia hair
- (f/i) = k = 100: stadia interval factor
- C = (c + f) approximately 0, Stadia constant

- D = KI + C = 100 I, for horizontal sight

H = 100 I cos2(α) V = 100 I sin(α) cos(α)

- Early types:
- Transmit light, measure up to 25 miles
- Transmit microwaves, measure up to 50 miles

- Classification of EDM:
- Electro-optical: laser or infra red reflected from passive prism or surfaces, the US has installed a prism on the moon.
- Microwave: two positive units, GPS replaced them for most engineering applications such as hydrographic surveys

Electromagnetic Spectrum

- We only measure the phase angle shift (change) , different signals of wave length: 10 100 1000 10,000 m are sent. Each fraction provides a digit(s).
- (Phase shift / 360)*wave length = non complete cycle length.
- Example: how a distance 3485.123 is measured.

Electronic Distance MeasurementEDM

The Idea:

To measure the distance between two points (A) and (B) the EDM on point (A) sends electromagnetic waves. The waves received at (B) are reflected back or resent to (A) by a device on (B).

- Knowing the speed of electromagnetic waves in the air, the EDM computes the distance by measuring the time difference or the shift of the wave phase angle (will be explained in details later).

Concept of the fraction

90

0

180

270

Phase Angle

Assume that = 2 m

If 1 = 80, it corresponds to a distance = (80/360) * = 0.44 m

If 2 = 135, it corresponds to a distance = (135/360) * = 0.75 m

If 3 = 240, it corresponds to a distance = (80/360) * = 1.33 m

- Distance = Velocity * Time = ((N *) + ) / 2
Where is a fraction of wave length = (/360) *

N is the number of full cycles, ambiguity?

Since is divided by 2, so is , we call /2 “effective wave length”

EDM Mounted on a theodolite

A

B

C

The angle between the rays A and C is double the angle between the two mirrors = 2 *90 = 180

Notice that the objects will look upside down, notice the box at the tail of the arrow

Aiming at a prism through the telescope of a total station in a zoo!

Reflectors (Prisms)

Fully rotating prism

Prism and sighting target

Pole and bipod

- A reflector might include a single prism or a group of prisms
- Reflectors can be a simple reflecting paper-sticker, they are called sheet-prisms, paper prism, or reflective sheeting. Very instrumental in construction sites and deformation monitoring of structures.

- 3mm, and 3ppm is the most common.
- Estimated error in distance =
Ei2 + er2 + ec2 + (ppm X D)2

- Where: Ei and er are the centering errors of the instrument and the reflector, ec is the constant error of the EDM, and PPM is the scalar error of the EDM
Example 6-5 page 156: if the estimates errors of centering the instrument and target were ± 3mm and ± 5mm respectively, the EDM had a specified error of ± (2mm +2ppm) what is the estimated error in measuring 827.329 m?

Answer: ± 6.4 mm

- Prism constant consideration.
- Total station Vs EDM.
- Data collectors.
- Prismless EDM: up to 100 m, can reach hard places, figure 6-14.

Handheld laser measuring devices