- 927 Views
- Uploaded on

Download Presentation
## PowerPoint Slideshow about 'Dynamic Analysis of First Order Instruments' - Patman

**An Image/Link below is provided (as is) to download presentation**

Download Policy: Content on the Website is provided to you AS IS for your information and personal use and may not be sold / licensed / shared on other websites without getting consent from its author.While downloading, if for some reason you are not able to download a presentation, the publisher may have deleted the file from their server.

- - - - - - - - - - - - - - - - - - - - - - - - - - E N D - - - - - - - - - - - - - - - - - - - - - - - - - -

Presentation Transcript

### Dynamic Analysis of First Order Instruments

P M V Subbarao

Professor

Mechanical Engineering Department

Capability to Reach Exact Equilibrium ……

First Order Instruments

A first order linear instrument has an output which is given by a non-homogeneous first order linear differential equation

- In these instruments there is a time delay in their response to changes of input.
- The time constant t is a measure of the time delay.
- Thermometers for measuring temperature are first-order instruments.

The time constant of a measurement of temperature is determined by the thermal capacity of the thermometer and the thermal contact between the thermometer and the body whose temperature is being measured.

- A cup anemometer for measuring wind speed is also a first order instrument.
- The time constant depends on the anemometer\'s moment of inertia.

First‐order Instrument Step Response

b0

The complex function F(s) must be decomposed into partial fractions in order to use the tables of correspondences. This gives

Thermometer: A First Order Instrument

Conservation of Energy during a time dt

Heat in – heat out = Change in energy

of thermometer

Assume no losses from the stem.

Heat in = Change in energy

of thermometer

Step Response of Thermometers

Time constant

Response of Thermometers: Periodic Loading

If the input is a sine-wave, the output response is quite different;

but again, it will be found that there is a general solution for all situations of this kind.

U-tube Manometer : A Second Order System

- The pressure to be measured is that of a system that involves a fluid (liquid or a gas) different from the manometer liquid.
- Let the density of the fluid whose pressure being measured be ρf and that of the manometer liquid be ρm.
- Equilibrium of the manometer liquid requires that there be the same force in the two limbs across the plane AA.
- We then have

p

patm

This may be rearranged to read

Dynamic response of a U tube manometer

h

- The manometer liquid is assumed to be incompressible the total length of the liquid column remains fixed at L.
- Assume that the manometer is initially in the equilibrium position.
- The pressure difference Δp is suddenly applied across it.
- The liquid column will move during time t > 0.

The forces that are acting on the length L of the manometer liquid are:

Force disturbing the equilibrium

Inertial Force

Forces opposing the change:

a. Weight of column of liquid

b. Fluid friction due to viscosity of the liquid :

- The velocity of the liquid column is expected to be small and the laminar assumption is thus valid.
- The viscous force opposing the motion is calculated based on the assumption of fully developed Hagen-Poiseuelle flow.

The fricitional pressure drop

Second Order System

The essential parameters

The static sensitivity:

The dimensionless damping ratio:

The Natural Frequency:

The transfer function is parameterized in terms of ζ and ωn.

- The value of ωn doesn’t qualitatively change the system response.
- There are three important cases—withqualitatively different system behavior—as ζ varies.
- The three cases are called:
- Over Damped System (ζ >1)
- Critically Damped System (ζ =1)
- Under Damped System (ζ <1)

t

Download Presentation

Connecting to Server..